How to clean Tidalflux electromagnetic meters for partially filled pipes?
asked 3 February 2016 asked by canardo26 (330 points)
Hi, You should ask Krohne, but should be okay as long as you reduce the pressure. Cheers
Hello, I think it is best to avoid a vacuum truck because the capacitive level sensor won't like it. Normally Khrone recommends having a dedicated opening either in front of or behind the sensor so it can be cleaned manually.
I second Tonino's response You should use some kind of plugged opening. The product specs refer to the following: Provide openings for inspection and cleaning upstream of the TIDALFLUX in order to periodically check the inner wall for accumulated deposits and to clean it if necessary using a water jet with pressure of no more than 10 bar.