Chlorine residual differences
Usually when a CL17 is reading high i Check to see what the reference number is to see if the colorometer cell is dirty. Next i will check the flow. When a unit reads too low it means there is too much flow and it dilutes the reagents too much. when it reads high it means there is not enough flow and the concentration is to high and reads high.